Author: Noyafa–POE Tester
Matters needing attention in the testing process of the cable fault tester The cable fault tester manufacturer is used to test the open circuit, short circuit, grounding, low resistance, high resistance flashover and high resistance leakage faults of power cables, as well as the same Axis communication cables and cable fault testers need to pay attention during the test cable fault tester manufacturer cable fault tester is used for power cable open circuit, short circuit, grounding, low resistance, high resistance flashover and high resistance test, and the ultimate test of open circuit and short circuit faults of coaxial communication cables and local telephone cables. It can also test the cable path, burial depth, and the speed of the radio wave, and verify the cable length, etc., and can establish a cable file for daily maintenance and management. The cable fault tester adopts the principle of time domain reflection, transmits a series of electric pulses to the cable under test, and receives the reflected pulses caused by the impedance change in the cable, and then according to the propagation speed of the electric wave in the cable and the two reflections The time represented by the characteristic inflection point of the wave can measure the distance from the fault point to the test terminal.
Precautions during the use of the cable fault tester manufacturer: 1. When testing by the pulse method, pay attention to throwing off all the equipment in the office and measure on the outer line. 2. When using the flashover method to test, the trigger working mode switch must be set to“Flashover”Location. 3. Pay attention to personal safety and equipment safety when using the direct flash method or the dash flash method for testing.
The ground wire must be connected. 4. After the flashover test is completed, cut off the power supply, remove the connecting wire between the instrument and the high-voltage test device, and then discharge the stored charge of the high-voltage capacitor and cable. When discharging, a current limiting resistor R should be added to limit the discharge current to discharge the current slowly. After the voltage on the capacitor is reduced, the resistance in the discharge circuit to the ground is zero, and the instantaneous discharge current can be as high as several hundred amperes, which will cause serious damage. equipment or personal accident.
5. During the test of the direct flash method, attention must be paid to monitoring the leakage current of the fault. If the current suddenly increases and the fault flashover does not appear, the test voltage should be lowered immediately and the flash-over method should be used instead. The smooth flow of the cable fault tester system and the transmission of power depend on the normal operation of the cable line. Once there is an obstacle in the line, it will cause interruption of communication and power. If the fault cannot be detected in time and eliminated quickly, it will cause great economic loss and adverse social impact.
Therefore, maintenance personnel should pay attention to ensure the normal operation of the cable fault tester during the work process. Cable fault tester manufacturer Shenzhen Noyafa Electronic Technology Co., Ltd.
Copyright © 2022 SHENZHEN NOYAFA ELECTRONIC CO.,LIMITED - www.noyafa.net All Rights Reserved.